KAPA Library Quantification Kits contain KAPA SYBR FAST DNA Polymerase, which was engineered through our directed evolution technology to amplify diverse DNA fragments with similar efficiency. This is important for the amplification of heterogeneous populations such as NGS libraries. Library quantification kits containing wild-type DNA polymerases only count “easy” library molecules.*
View data showing efficient amplification and quantification of diverse libraries by KAPA SYBR FAST DNA Polymerase compared to other options.
KAPA SYBR FAST exhibits high-efficiency amplification and library quantification.
(A) Amplification efficiencies achieved with KAPA SYBR FAST for ten diverse amplicons, plotted against GC content. All ten amplicons were amplified with similar efficiency (within the optimal range of 95% – 105%). This confirms that KAPA SYBR FAST is the ideal enzyme for the amplification of heterogenous populations of DNA fragments, such as NGS libraries.
(B) DNA fragment size distributions for an NGS library before (grey fill) and after qPCR amplification with KAPA SYBR FAST and two competitor kits containing wild-type Taq polymerase. Reactions were performed with the following cycling protocol: 95°C for 10 min, followed by 40 cycles of 95°C for 10 sec, and 60°C for 45 sec. The KAPA SYBR FAST quantification product is representative of the template DNA (NGS library), whereas the quantification products generated with wild-type enzymes are not.
Generation of standard curve and quantification of library concentration
Six pre-diluted DNA Standards and appropriately diluted NGS libraries are amplified using platform-specific qPCR primers that target adapter sequences. The average Cq value for each DNA Standard is plotted against its known concentration to generate a standard curve. The standard curve is used to convert the average Cq values for diluted libraries to concentration, from which the working concentration of each library is calculated.
Standard protocols for all three major commercial Next Generation Sequencing (NGS) platforms employ unreliable, laborious, and costly methods for quantifying library DNA molecules prior to clonal amplification of sequencing templates. Accurate quantification of bona fide PCR-competent sequencing templates is crucial for reliable clonal amplification via bridge PCR (bPCR; “cluster amplification”) or emulsion PCR (emPCR) – underestimation results in non-clonality and/or over-clustering, while overestimation leads to poor yields of clusters or template-carrying beads.
Most standard methods for quantifying NGS libraries have a number of important disadvantages. First, electrophoresis and spectrophotometry measure total nucleic acid concentrations, whereas optimal cluster density or template-to-bead ratio depends on the appropriate input concentration of PCR-amplifiable DNA templates. Since the proportion of amplifiable DNA molecules in a library may vary with each sample, expensive and time- consuming titrations are required. Second, these methods have low sensitivity, consuming nanograms of precious samples, or about 1000 times more molecules than are required for sequencing. Finally, electrophoresis and spectrophotometry are not suited to high-throughput of samples, requiring laborious and error-prone manual liquid handling.
We believe that qPCR quantification of libraries is the best approach for minimizing variability in cluster density or bead enrichment. Aside from errors in data collection, processing, and/or analysis there are three possible explanations for such differences:
A single KAPA Library Quantification Kit is sufficient for the quantification of ~30 libraries (following our recommended protocols in a 96-well format). However, more libraries can be quantified using:
Many variables will greatly affect the ultimate number of libraries that can be quantified per kit. Below are a few guidelines based on the details of our recommended protocols:
|Reliability of Data|
|3||1||118||287||Acceptable, with increased risk of repeats|
|2||2||95||218||Acceptable, with increased risk of repeats|
|2||1||190||437||Acceptable, with increased risk of repeats|
*We recommend triplicate reactions for the DNA standards and for the library samples; however, some users may feel sufficiently confident, especially after some experience with the kit, to do away with some replicates of the standards and/or the samples. This would make it difficult or impossible to troubleshoot any unexpected results and the quantification accuracy may suffer; on the other hand, many more libraries could be quantified with the same amount of reagent.
**We recommend that you perform three serial dilutions of each library sample to ensure that at least one dilution will fall within the upper limit of the range of the assay. This also allows calculation of the qPCR efficiency for that particular library sample. Thus, it is possible that with experience and/or with very standardized workflows, you may feel confident to stop doing these serial dilutions, in which case the number of qPCR reactions required for each quantification will be reduced.
***If you quantify a single library sample at a time, you will be able to quantify six libraries per kit of DNA standards. Each quantification experiment will use up 6 x 3 qPCR reactions for the standards plus 4 x 3 qPCR reactions for the library sample = 30 qPCR reactions in total. Since each Kit contains 5 mL KAPA SYBR FAST 2X qPCR Master Mix (sufficient for 500 x 20 μL reactions), you will have 500 μL – (6 x 30 μL) = 320 qPCR reactions left over when you have finished the DNA Standards. In this case, additional DNA Standards can be purchased separately, so that the leftover qPCR reagent is not wasted.
****Users may choose to run 10 μL volumes (especially in 384-well formats). The volume of DNA Standard added to each reaction remains 4 μL, regardless of the reaction volume.
Inaccurate liquid handling is the most common reason for variability. Accurate quantification requires careful pipetting. For liquid handling systems, consult the relevant user manual. Alternatively, apply the following when using a non-automated device:
Contamination of standards and libraries can also lead to error in the assay. We recommend:
Due to the design specifications and broad dynamic range of the KAPA Library Quantification assay, DNA Standard 1 has a higher copy number than samples that are typically analyzed by a qPCR, particularly in relative quantification assays. As a result, Standard 1 (and any samples that were not sufficiently diluted) may return a very early Cq score and/or an increasing fluorescent signal during the automated baseline determination performed by the qPCR instrument. Many instruments use the early cycles to calculate and set the baseline subtraction/ correction, and samples that are changing significantly in fluorescence during this phase can dramatically affect this process (since there is no stable baseline).
It is usually possible to tell whether this is an issue by examining the amplification plots. If the corrected/processed amplification plot begins well below the baseline of samples crossing the Cq later, then it is likely that baseline determination/subtraction was not successful for that plot. Another useful way to judge whether an early Cq value has created problems during data analysis is to confirm that the expected spacing between consecutive standards is ~3.32 cycles (the standards represent a 10-fold dilution series). Similarly, consecutive two-fold dilutions of a given sample should cross the Cq ~1 cycle apart.
If you feel that the first standard and/or some sample dilutions have Cq scores that are too early, then simply exclude that standard and those sample dilutions from the analysis. This is most easily done by de-selecting them in the qPCR analysis software. If you experience this issue regularly with your instrument/settings/workflow, then you may choose to omit the first DNA standard from all future assays, and you might want to consider implementing a larger up-front dilution of your library samples in your standard workflow. If the first standard is omitted, ensure that the library Cq score falls within the dynamic range of Standards 2-6.
When examining the data from the DNA standards, one should look at two things first:
Next-generation sequencing libraries are generally complex, comprising a very wide diversity of DNA fragments.While KAPA SYBR FAST qPCR reagents are generally capable of extremely fast thermocycling, we conservatively recommend relatively long denaturation and annealing/extension times for library quantification in order to accommodate the diversity of templates in a typical library sample. The recommended qPCR protocol for library quantification consists of an initial denaturation step at 95°C for 5 minutes followed by 35 cycles of denaturation at 95°C for 30 seconds and combined annealing/ extension at 60°C for 45 sec. If the average library fragment size is >700 bp, increase annealing/extension time to 90 seconds.
A melt curve analysis is typically included at the end of SYBR Green I-based qPCR assays to provide information about the specificity of the reaction. In library quantification melt curve analysis is less relevant, as the template being amplified is not comprised of a single DNA fragment of defined length and GC content, but is a heterogenous population of molecules, with a range of lengths and GC contents. Unlike the melt curve for a single amplicon, the melt curve for an Illumina library is a “composite” analysis, representing an average across the entire population of library fragments. It is also important to keep in mind that library samples are expected to reach their quantification cycle (Cq) relatively early, whereas the melt curve analysis is only performed after 35 cycles, i.e. at a time when substrate depletion has occurred and amplification artifacts that are no longer representative of the original template have been generated. This complicates the interpretation of melt curves generated in library quantification assays.
Nevertheless, we have found melt curve analysis useful for the identification of carry-over adapter dimer in Illumina libraries. Adapter dimers, formed during library construction with full-length adapters, are very efficiently amplified in the KAPA Library Quantification assay, and will lead to an overestimation of library concentration if present in significant quantities. A small second peak of higher temperature to the right of the main library peak will indicate adapter dimer. Please note that melt curves for the DNA Standard included in KAPA Library Quantification Kits for Illumina platforms display a very characteristic double peak. This is the result of differential local melting in the 452 bp linear standard.
Primer P1: 5′-AAT GAT ACG GCG ACC ACC GA-3′
Primer P2: 5′-CAA GCA GAA GAC GGC ATA CGA-3′
Primer IT A: 5′-CCA TCT CAT CCC TGC GTG TC-3′
Primer IT trP1: 5′-CCT CTC TAT GGG CAG TCG GTG AT-3′
Applied Biosystems SOLiD
Primer P1: 5′ – CCA CTA CGC CTC CGC TTT CCT CTC TAT G – 3′
Primer P2: 5′ – CTG CCC CGG GTT CCT CAT TCT – 3’
The DNA Standards used in our Library Quantification Kits are not sequencing libraries, but rather a defined, pure, linear, dsDNA amplicon for each set of DNA Standards.This allows us to rigorously validate their efficiency and reproducibility for use as qPCR amplification standards. Before accepting a newly manufactured lot into our inventory, we use a stringent qPCR assay to compare each new lot of KAPA Library Quantification DNA standards to a reference set of standards during manufacturing and quality control.We compare Cq scores for each standard in a newly manufactured set with Cq scores in a reference set of standards, and we ensure that each standard lies within 0.1 Cq of the respective reference standard, and that the resulting standard curve essentially lies on top of the reference standard curve (minimal deviations in y-intercepts and slopes).
KAPA Library Quantification Kits are shipped on dry ice or ice packs, depending on the country of destination. Upon receipt, store the entire kit at -20°C in a constant-temperature freezer. When stored under these conditions and handled correctly, all kit components will retain full activity for at least six months from the date of receipt.
All components of the KAPA Library Quantification Kits – as well as the combined KAPA SYBR FAST/Primer Premix solution – are stable through more than 30 freeze/thaw cycles. We therefore recommend that all reagents are stored in the dark at -20°C when not in use. Nevertheless, these reagents are stable in the dark at 4°C for at least one week, and may be stored in this state for short-term use, provided that they are not contaminated with microbes and/or nucleases.
For certain real-time cyclers, the presence of ROX reference dye compensates for non-PCR-related variations in fluorescence detection. The fluorescence level of ROX reference dye does not change significantly during the course of real-time PCR, but provides a stable baseline against which PCR-related fluorescent signals are normalized. Thus, ROX dye compensates for differences in fluorescence detection between wells due to slight variations in reaction volume or differences in well position. The presence of ROX dye in the master mix does not interfere with real-time PCR on any instrument, since the dye is not involved in the reaction and has an emission spectrum different from that of SYBR Green I. If using the kit with Universal qPCR Master Mix and will always only use ROX High or ROX low, the entire 200 µL of appropriate ROX solution may be added to the qPCR Master Mix when opened. If ROX will be added individually, add 0.4 μL of 50X ROX High/Low to each assay.